RPH Keyhole Dex Sling
$412.00 AUD – $478.00 AUD
- Important Information
- Full body keyhole plate Sling
- Split legs
- Built-in Keyhole Plate Safety Straps
The RPH Keyhole Dex Sling is a full body keyhole plate sling. It has head support with split leg sections to allow easier fitting, and is available in a variety of materials (see the FAQ tab).
This Sling comes with keyhole plates to fit onto Tilt Frame Hoists that have mushroom studs to attach the Keyhole Plates to.
We incorporate special ‘Safety Straps’ with our slings as an extra safety measure.
Safe Working Load: 220kg / 485lbs (Weight Tested To 550kg / 1212lbs, exceeding ISO 10535:2011)
Attachment Method: Keyhole plate tilt frame hoist
|Item #||Size||Height||Shoulder Width||Hip Width||Webbing Colour|
|771X||Extra Small||73cm / 28.5"||42cm / 16.5"||63cm / 25"||Brown|
|771||Small||82cm / 28.5"||62cm / 24.5"||80cm / 31.5"||Red|
|773||Medium||90cm / 35.5"||66cm / 26"||83cm / 32.5"||Yellow|
|770||Large||90cm / 35.5"||92cm / 36"||94cm / 35.5"||Green|
|772||Extra Large||105cm / 41.5"||96cm / 38"||114cm / 45"||Blue|
Special sizes made on request.
||Nylon||This is called a solid material, as it is not a Mesh or Netting material that has holes in the material. The Nylon has some slipperiness, to make it easier to slide down behind a patient's back, when they are sitting in a chair.|
||Polyester||This is a soft and comfortable Mesh with small holes. The holes can be cooler for the patient. Sometimes this material will be used for showering or in a bath.|
||Polyester||Like the Mesh, but with bigger holes. Used more for bathing or showering, as more water will drain from the large holes and it is possible to pat dry the patient with a towel after showering.|
||Polyester||A comfortable, solid material, like the Nylon material, but not as slippery.|
|All Day Sling Material
||Nylon||A strong, thin material with a Rip Stop thread in the material, to help reduce or prevent ripping. As the material is breathable the Slings can be left under the patients sitting in chairs. This reduces the need to remove the Sling after the patient has been hoisted into a chair, and then replace the Sling to pick them up again.|
Please select the most appropriate material and sling design for the patient, but remember, if the patient’s condition changes, it may be necessary to use a different material and design of sling.
See the FAQ tab for more information.
Wash up to 80°C (176°F). Air or tumble dry on a cool setting, ensuring the drum is cool. Ensure the product is dry before using or storing. We recommend using a Pelican Washing Bag to help contain the straps.
Attaching Brand New Keyhole Plate Slings to a Hoist: View our video guidance here.
General Keyhole Plate Sling Information: Click here for general information, including filing down Keyhole Plates which are too tight.
Warning: Inspect slings before every use. If any damage is found the Sling must be withdrawn from service and repaired, if possible.
Disposal: When the Sling is ready for disposal, we suggest the Lifting Straps are cut off, so it cannot be used with another patient.
Sling & Hoist Compatibility: Click here for more information.
Sling Life Expectancy: Click here for more information.
What the International Standard ISO10535:2011 requires for Period Sling Inspections and Sling Labelling: Click here for more information.
Need Simple Sling Inspections? For the 6 Monthly Inspections we recommend using the SlingSpector platform and app to conduct safe, efficient sling inspections, with consistency across an entire organisation.
Do I need to do Sling inspections?
Yes - see the Important Information tab for more information.
What are the Sling inspection requirements?
See the Important Information tab for more information. We've paraphrased the International Standard 10535:2011 under licence, detailing exactly what is required.
Where can I find the inspection checklists?
See the Important Information tab for more information. We provide the following checklists:
- Sling Inspection Before Use
- 6 Monthly Thorough Sling Inspection