Hammock with Hole Sling
$227.00 AUD – $313.00 AUD
- Important Information
- Good hygiene access
- Hammock style
This Sling is available in a variety of materials (see the Technical tab). It has been more popular with some double amputees or contracted patients in fall-out chairs as they may find it more comfortable than the split leg Sling design. Patients sitting in fall-out chairs can also be rolled to place the Sling under them.
To put on the patient, it is best to lay the folded Sling on the bed next to the patient and then roll the patient onto the Sling, similar to placing a draw sheet under a patient. The hole is provided for toileting and once fitted, it may be best to leave the patient sitting in the Sling to avoid refitting once the patient is sitting down in a chair. A central vertical line of stitching indicates the middle of the Sling. This should be lined up with the patient’s spine so the Sling is evenly fitted.
This Sling is also available in a strong netting or mesh material. The large holes of the netting Sling allow water to drain away after bathing. It also allows Carers to pat the patient dry while in the Sling. The smaller holes in the mesh material allow for bathing and both Slings give ventilation when the Slings are left under the patients.
A similar Sling is the Hammock with Chair Pad Sling. This Sling has a non-slip backing and a Chair Pad is sewn to the inside. The non-slip backing keeps the Sling in position in the chair. The Chair Pad gives extra padded insulation from vinyl covers on the chairs. The Sling can be left under the patient reducing the manual handling and discomfort for the patient.
Item # 570
Safe Working Load: 200kg / 440lbs (Weight Tested To 700kg / 1540lbs, exceeding ISO 10535:2011)
Attachment Method: Metal hook coathanger type hoists
81cm / 32″
90cm / 35.5″
93cm / 36.5″
122cm / 48″
Special sizes made on request.
|Standard White||Nylon||This is called a solid material, as it is not a Mesh or Netting material that has holes in the material. The Nylon has some slipperiness, to make it easier to slide down behind a patient's back, when they are sitting in a chair.|
|Blue Mesh||Polyester||This is a soft and comfortable Mesh with small holes. The holes can be cooler for the patient. Sometimes this material will be used for showering or in a bath.|
|Polyester Blue||Polyester||A comfortable, solid material, like the Nylon material, but not as slippery.|
|All Day Sling Material||Nylon||A strong, thin material with a Rip Stop thread in the material, to help reduce or prevent ripping. As the material is breathable the Slings can be left under the patients sitting in chairs. This reduces the need to remove the Sling after the patient has been hoisted into a chair, and then replace the Sling to pick them up again.|
Please select the most appropriate material and sling design for the patient, but remember, if the patient’s condition changes, it may be necessary to use a different material and design of sling. See the FAQ tab for more information.
Wash up to 80°C (176°F). Air or tumble dry on a cool setting, ensuring the drum is cool. Ensure the product is dry before using or storing. We recommend using a Pelican Washing Bag to help contain the straps.
Warning: Inspect slings before every use. If any damage is found the Sling must be withdrawn from service and repaired, if possible.
Warning: If you are leaving a sling made in the All Day Sling Material option under a person, always ensure that there will be no adverse pressure care issues. This includes checking that the attachment method (e.g. webbing loops, keyhole plates, etc.), are not positioned in such a way that may hurt the person. The actual material is very thin and it should not cause a pressure problem, but patients’ skin integrity should be regularly checked, just in case the sling is causing a pressure problem. If there are any early signs of pressure problems, discontinue using this design of sling.
Disposal: When the Sling is ready for disposal, we suggest the Lifting Straps are cut off, so it cannot be used with another patient.
Sling & Hoist Compatibility: Click here for more information.
Sling Life Expectancy: Click here for more information.
What the International Standard ISO10535:2011 requires for Period Sling Inspections and Sling Labelling: Click here for more information.
6 Monthly Thorough Sling Inspection Checklist: We adopt the SlingSpector checklist. Click here to download.We recommend using colour-coded Sling Inspection Tags to help reduce staff's time when searching for Slings which need inspecting.
Need Simple Sling Inspections?: For the 6 Monthly Inspections we recommend using the SlingSpector platform and app to conduct safe, efficient sling inspections, with consistency across an entire organisation.
Do I need to do Sling inspections? Yes - see the Important Information tab for more information.
What are the Sling inspection requirements? See the Important Information tab for more information. We've paraphrased the International Standard 10535:2011 under licence, detailing exactly what is required.
Where can I find the inspection checklists? See the Important Information tab for more information. We provide the following checklists:
- Sling Inspection Before Use
- 6 Monthly Thorough Sling Inspection